Message 3 of 8. I want labview to basically be able to read the voltage values and plot it vs. Message 6 of 8. When you get problems with your VI, come back here, attach your VI with description what you tried, and what is your goal. Check to make sure it is in the remote mode. Loop until done or until an error or until stopped by the user.
|Date Added:||9 January 2015|
|File Size:||10.12 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Keithley Electrometer – Instrument Driver – National Instruments
Message 8 of 8. I do not like inputs on the right and outputs on the left. This process may take up to one U.
Message 4 of If the error occurs on the first write, then the instrument is probably not responding. I now know enough to know that I have no clue about anything labbiew all. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
I am rewriting all the VIs I need for other Keithley instruments. Message 3 of We apologize for any inconvenience.
The scan may or may not list which instrument is which. Message 6 of 8.
Loop until done or until an error or until stopped by the user. My support system ensures that I don’t look totally incompetent.
Instrument Control (GPIB, Serial, VISA, IVI)
To measure low current by using electrometer When you get problems with your VI, come back here, attach your VI with description what you tried, and what is your goal. Here is one example VI, try to use it C: Most Active Software Boards: Message 9 of If your Keithley devices have RS ports no ideathen you could also work via serial port, would be cheaper.
I do not have the instruments so I cannot actually test it. Message 5 of 8. Yes, You need a loop. Message 4 of 8.
Keithley – Discussion Forums – National Instruments
If you have kkeithley questions, please contact us at Try allowing several seconds. Message 10 of GPIB error 6 is a timeout error. You have two different controls wired to two of the functions and nothing wired to the others.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. The Tektronix website merged Keithley one in went crazy, they do not let me download the driver:.
Im doing a current sweep from 0mA to. I looked at your VI and downloaded the Keithley drivers. Write in low current. Message 7 pabview 8.